One-jet water meter "Pulsar" with visual reading is designed to measure the volume of hot and cold drinking water according to SanPiN 188.8.131.524-01 and network water flowing through the pipeline in cold and hot water supply systems. The data is displayed on the front panel of the computing mechanism. To read them, direct access to the meter is required.
Information on the testimony is collected by residents, employees of utilities or the management company by door-to-door round-trip. The advantages of this modification of the water meter are high reliability and affordable price.
|Nominal pressure MPa||1,6|
|Permissible error in the range Qt ≤ Q ≤ Qmax, %||2|
|Permissible error in the range Qmin ≤ Q ≤ Qt, %||5|
|Water temperature||+5 °С cold water, +90 °С hot water|
|Number in the Register of Measuring Instruments of the Russian Federation||63458—16|
|Features of the||Ду15||Ду15||Ду15||Ду20||Ду20|
|Water consumption, m3/h|
|— least, Qmin||0,06||0,03||0,015||0,1||0,05|
|— transition, Qt||0,15||0,12||0,022||0,25||0,2|
|— nominal, Qn||1,5||1,5||1,5||2,5||2,5|
|— the greatest, Qmax||3,0||3,0||3,0||5,0||5,0|